Verify that at least 3 bars are displayed.
CAM display should clear to one or zero bars within two
CAM is ready for operation in H Mode.
through (7) using the G vapor shroud of confidence tester. After
steps are performed for the G Mode, the CAM will be ready for
operation in G Mode.
c. Operation of CAM.
Turn CAM on.
Perform self test (performed in H Mode).
Place CAM in H Mode.
Hold CAM 1/2 inch from the object being monitored.
(6) To change from H Mode to G Mode, press G H Mode
pushbutton switch (1); verify display (2) indicates G Mode. WAIT may
be displayed for several seconds. When WAIT has disappeared,